The LEGENDplex™ SARS-CoV-2 Serological IgG Panel (3-plex) is a bead-based multiplex assay panel, using fluorescence-encoded beads suitable for use on various flow cytometers. It allows for simultaneous semi-quantification of IgG titer of the 3 key COVID-19 coronavirus derived proteins: spike 1 (S1), receptor binding domain (RBD) and nucleocapsid (N). This assay panel provides higher detection sensitivity and broader dynamic range than traditional ELISA methods. The panel has been tested for use on serum and plasma samples.
The LEGENDplex™ SARS-CoV-2 Serological IgG Panel is designed to allow flexible customization within the panel. Please visit www.biolegend.com/legendplex for more information on panel design and how to mix and match within the panel.
This assay is for research use only.
- Kit Contents
- Setup Beads 1: FITC Beads (1 mL)
- Setup Beads 2: PE Beads (1 mL)
- Setup Beads 3: Raw Beads (2 mL)
- SARS-CoV-2 Serological IgG Panel Premixed Beads (3.5 mL)
- SARS-CoV-2 Serological Panel IgG Detection Antibodies (3.5 mL)
- SARS-CoV-2 Serological Panel Standard, Lyophilized (1 vial)
- LEGENDplex™ SA-PE (3.5 mL)
- LEGENDplex™ Assay Buffer (25 mL)
- LEGENDplex™ Wash Buffer, 20X (25 mL)
- Plate Sealers (4 sheets)
- V-bottom Plate
Learn more about LEGENDplex™ at biolegend.com/legendplex
Download the LEGENDplex™ software here.
- Product Citations
- Materials Not Included
- Flow Cytometer
- Pipettes and Tips
- Reagent Reservoirs for Multichannel Pipettes
- Polypropylene Microfuge Tubes
- Vortex Mixer
- Aluminum Foil
- Absorbent Pads or Paper Towels
- Plate Shaker
- Tabletop Centrifuges
- A Vacuum Filtration Unit and a Vacuum Source (if using filter plates)
- Centrifuge with a swinging bucket adaptor for microtiter plates (if using V-bottom plates)
- 1.1 mL polypropylene micro FACS tubes, in 96-tube rack
- If I don't have a vacuum, how do I remove the liquid from my plate?
If you do not have a vacuum, the assay should be run in a V-bottom plate. After centrifugation using a swinging-bucket rotor with a plate adaptor, you can remove the liquid by flicking the plate quickly, dumping the contents into a sink, and patting it dry carefully on a stack of clean paper towels without losing the beads. Alternatively, you can remove the liquid by using a pipette.
- Should I perform the assay with the filter plates or with V-bottom plates?
Filter plates or V-bottom plates have been included in some kits for your convenience. A vacuum filtration unit is required to work with the filter plates. However, if you don’t have access to a vacuum manifold or if you prefer, then you can use the V-bottom plates and follow the recommended assay protocols for the type of plates you choose. All plates should be made from low binding polypropylene. Polystyrene ELISA or cell culture plates should not be used.
- After I finish the staining process, how long can I wait before reading my LEGENDplex™ samples?
The samples can be kept overnight at 4°C while being protected from exposure to light and be read the next day. There may be a decrease in signal, but overall, the assay results should not be affected. Storing the samples for extended periods of time is not recommended, as it could lead to further reductions in signal.
- What is the shelf life of LEGENDplex™ kits?
LEGENDplex™ kits are guaranteed for 6 months from the date of receipt, but may have a shelf life of up to 2 years from the date of manufacture.
- Is special software required for data analysis?
Typically flow cytometers generate output files in FCS format (e.g. FCS 2.0, 3.0, or 3.1) and in some cases in list mode file format (LMD). Other software may be available to analyze FCS files. Data generated using LEGENDplex™ kits can be analyzed using the freely available LEGENDplex™ data analysis software. Please check our website for the most updated versions of the software.