- Regulatory Status
- Other Names
- human IgA, human IgG
- Ave. Rating
- Submit a Review
- Product Citations
The LEGENDplex™ SARS-CoV-2 Serological Panel Standard contains 3 antibodies including anti-SARS-CoV-2 Spike S1/RBD Human IgG, anti-SARS-CoV-2 Nucleocapsid Human IgG, and anti-SARS-CoV-2 Spike S1/RBD Human IgA. This Standard cocktail is intended for use with the following Mix and Match assay reagents:
SARS-CoV-2 Serological Panels
741135 LEGENDplex™ SARS-CoV-2 S1 Spike Protein Capture Bead B2, 13X
741136 LEGENDplex™ SARS-CoV-2 S Protein RBD Capture Bead B5, 13X
741137 LEGENDplex™ SARS-CoV-2 Nucleocapsid Capture Bead B4, 13X
740375 LEGENDplex™ Buffer Set D
741140 LEGENDplex™ SARS-CoV-2 Serological IgA Panel Detection Abs
741133 LEGENDplex™ SARS-CoV-2 Serological IgG Panel Detection Abs
740377 LEGENDplex™ Filter Plate
740379 LEGENDplex™ V-bottom Plate
For product manuals for these panels, please visit the Mix and Match tab on the LEGENDplex™ page.
- If I don't have a vacuum, how do I remove the liquid from my plate?
If you do not have a vacuum, the assay should be run in a V-bottom plate. After centrifugation using a swinging-bucket rotor with a plate adaptor, you can remove the liquid by flicking the plate quickly, dumping the contents into a sink, and patting it dry carefully on a stack of clean paper towels without losing the beads. Alternatively, you can remove the liquid by using a pipette.
- Should I perform the assay with the filter plates or with V-bottom plates?
Filter plates or V-bottom plates have been included in some kits for your convenience. A vacuum filtration unit is required to work with the filter plates. However, if you don’t have access to a vacuum manifold or if you prefer, then you can use the V-bottom plates and follow the recommended assay protocols for the type of plates you choose. All plates should be made from low binding polypropylene. Polystyrene ELISA or cell culture plates should not be used.
- After I finish the staining process, how long can I wait before reading my LEGENDplex™ samples?
The samples can be kept overnight at 4°C while being protected from exposure to light and be read the next day. There may be a decrease in signal, but overall, the assay results should not be affected. Storing the samples for extended periods of time is not recommended, as it could lead to further reductions in signal.
- What is the shelf life of LEGENDplex™ kits?
LEGENDplex™ kits are guaranteed for 6 months from the date of receipt, but may have a shelf life of up to 2 years from the date of manufacture.
- Is special software required for data analysis?
Typically flow cytometers generate output files in FCS format (e.g. FCS 2.0, 3.0, or 3.1) and in some cases in list mode file format (LMD). Other software may be available to analyze FCS files. Data generated using LEGENDplex™ kits can be analyzed using the freely available LEGENDplex™ data analysis software. Please check our website for the most updated versions of the software.